The DT-900 TechStrobe from Shimpo Instruments is the perfect tool for slow motion, or stop action stroboscopic inspection. The Shimpo TechStrobe is the only palm size unit to include a Phase shift feature. Phase shifting allows the TechStrobe operator to advance or retard the flash timing by degrees° of angle, without changing the actual flash rate (FPM) of the TechStrobe.Employing its unique Phase shifting feature, the TechStrobe “freezes” rotating or reciprocating machinery for visual inspection of covered gear teeth, shafts, oil/fuel/ink injector patterns, and print media; all without stopping production.
The TechStrobe is completely portable and operates up to 2.5 hours at 1600 FPM from its detachable, rechargeable battery. The ergonomically designed battery serves as the padded grip area. The battery is quickly detached and recharged while another battery is snapped into place.
Designed for right or left hand, all TechStrobe functions are intuitive and easy to use. Quality and Maintenance Professionals will appreciate the rotary and push button controls for internal flash (FPM/RPM) control, external triggering, tachometer mode and Phase shift control. The push buttons rapidly increase (x 2), or decrease (÷2) the flashes per minute (FPM) and change by degrees° the Flash phase.
Compact, impact resistant and rugged, with that balanced feeling for one-handed operation, the TechStrobe will quickly become one of the Quality and Maintenance Professionals’ favorite tools.
Rechargeable Replaceable Battery.
Long Battery Charge Life: Minimum of 2.5 hrs at 1600 FPM (flash per Minute).
Intuitive Control Layout: Simple to operate.
Rotary and Push Button Flash Control: Fast Up/Down push button, fine tune with rotary encoder.
Flash Phase Shift: Precise flash timing control (5 degree increments) push button or rotary control.
Tach Mode: Provide RPM display when external sensor is attached.
Flash Rate Range: 40 – 12,500 flashes per minute (FPM)
Flash Rate Accuracy: ±0.5 FPM or ± 0.01% of reading, whichever is greater
Flash Setting Resolution: 0.1 FPM for 100 to 9999 FPM, 1 FPM for 10,000 to 12,500 FPM
Tachometer Range: 40 – 59,000 RPM
Tachometer Accuracy: ±0.5 RPM or ±0.01% of reading, whichever is greater
Flash Tube: Xenon, 10 w, user replaceable
Flash Tube Life: 100 million flashes
Flash Duration: 9-15 sec
Light Power: 154 mJoules per flash
Display: 8 character by 2 line LCD, alphanumeric
Display Update: Continuous
Display Resolution: 0.1 FPM for 100 to 9999 FPM, 1 FPM for 10,000 to 12,500 FPM
Timebase: Crystal oscillator, 100 ppm accuracy
Battery Type: NiMH, rechargeable, removeable
Battery Capacity: 2.6 AmpHr
Battery Charge Time: 2-4 hours, using supplied AC adapter
Run Time Per Charge: 2.5 hours at 1600 FOM, 1.25 hours at 3200 FPM
Battery Charger AC Input: 115 VAC, 50/60 Hz
Controls: Power, multiply by 2, divide by 2, phase shift, external trigger
External Trigger Input: 0-5V TTL type via stereo phono jack
To provide the best experiences, we use technologies like cookies to store and/or access device information. Consenting to these technologies will allow us to process data such as browsing behavior or unique IDs on this site. Not consenting or withdrawing consent, may adversely affect certain features and functions.
The technical storage or access is strictly necessary for the legitimate purpose of enabling the use of a specific service explicitly requested by the subscriber or user, or for the sole purpose of carrying out the transmission of a communication over an electronic communications network.
The technical storage or access is necessary for the legitimate purpose of storing preferences that are not requested by the subscriber or user.
The technical storage or access that is used exclusively for statistical purposes.The technical storage or access that is used exclusively for anonymous statistical purposes. Without a subpoena, voluntary compliance on the part of your Internet Service Provider, or additional records from a third party, information stored or retrieved for this purpose alone cannot usually be used to identify you.
The technical storage or access is required to create user profiles to send advertising, or to track the user on a website or across several websites for similar marketing purposes.